Electronic Muscle Stimulator is designed for
ease of operation.
The S-3500 provides 3 pre-programmed selector settings which save
valuable treatment time. The S-3500 has a pulsed uniphasic waveform
and a fixed pulse duration on each of the settings to provide
for a comfortable and effective muscle stimulation and contraction,
including large muscle groups, (i.e. quadriceps and trapezius)
with a minimal amount of output current (0-25 milliamps in normal
use). Soken is backed by a one year warranty and years of customer
High Frequency: Variable pulses per second
(200 to 50pps.) dependent upon output level
for enhanced comfort.
High Frequency with Preset Ramp Duty Cycle:
per second (200 to 50pps) dependent upon output
level. Duty Cycle Mode #2: Output ramps up to set
intensity. Remains at set intensity "ON-Time" for 8
Output ramps down to 0; "OFF-Time" and remains
at 0 for 2 seconds; cycle then repeats.
Low Frequency: 2 pulses per second.
- 1 Year
- Solid State
Duration: 500m sec. /1000m sec.
- Power Supply:
4 size C-cell Alkaline batteries
unloaded output: 170 volts (± 10%)
loaded output (w/skin impedence) 105 volts
output is controlled by the OUTPUT dial calibrated
on a vernier scale of 0-100
of 3 pre-set treatment modes to include preset
ramp duty cycle at mode #2
- 0-30 minute
and Negative Polarity switch
pads-2 standard, 4-8 pad capability w / bifurcator
level light, ON-OFF power light
8½"W x 3¼ "H x 7¼"D
8.3cm x 18.5cm)
6 lbs. 4oz. (2.87Kg)
includes an instruction manual, a warranty card, 2 treatment pads,
1 dispersive pad, 1 lead wire, 1 dispersive lead wire, 2 Velcro
straps, & 4 C-cell batteries.
Federal Law (USA) restricts this device
to sale by or on the order of a qualified practitioner or qualified
physician. EMS devices should be kept out of the reach of children.
instruction manual for proper use of this device.
Muscle Stimulators (EMS) should not be applied where stimulation
in not desirable; for example:
- High fever,
tuberculosis, malignant tumor, serious injury.
expectorant bleeding, hematemesis.
sickness or injury.
with pacemakers or other heart conditions.
- Avoid use
in the areas of the brain and spinal column.
- The safety
of using EMS during pregnancy or over long periods of time is
should be used in the transthoracic application of EMS devices.